New Design Portable Dental X-ray Unit (tdh-hx01)
|FOB Unit Price:||US $800 US $600|
|Purchase Qty. (Pieces)||FOB Unit Price|
|Production Capacity:||500PCS Monthly|
|Transport Package:||Polywood Case|
|Payment Terms:||L/C, T/T, D/P, Western Union, Paypal, Money Gram|
- Model NO.: tdh-hx01
- Certification: RoHS, SGS, ISO, UL, CE
- Material: Metal
- Medical Devices Reg./Record No.: Gxzz20132551382
- Tube Voltage: 60kv
- Input Power: <600va
- Irradiation Diameter: 6cm
- Capacity: 60kv . 8mA
- Lifetime: 6 Years
- Specification: CE ISO approved
- HS Code: 90184990
- Applicable Departments: Orthodontic Department
- Type: X-Ray Equipment
- Medical Device Regulatory Type: Type 2
- Voltage: 110V-220V 50Hz
- Tube Current: 8mA
- Exposure Time: 0.2~4.0s
- Fuse: RF1-20-5A
- X-ray Area: 3.14X262(mm2)
- Trademark: TDH
- Origin: China
new design portable Dental X-ray Unit (tdh-hx01)
dental x-ray unit is an essential diagnostic equipment for dentist with features as following:
a). Simple structure: The x-ray generator, central controller and unit frame are integrated into one unit.
b). Good protection: The radiation is much lower than national protecting standard.
c). Digital circuit: The digital circuit makes exposure time accurately.
1.2 Main use
The unit is particularly provided for synthesis hospitals,cavity specialty hospital and dental clinic to take photo.
1.3 Environmental and working conditions
a. Condition of transportation and store
Environmental temperature: -20~70ºC
Atmospheric pressure: 50~106Kpa106Kpa
b. Conditions of operation
Environmental temperature: 10~40ºC
Atmospheric pressure: 70~
c. Power supply
Single phase AC 220±22V;50Hz±1Hz
1.4 Security and relative symbol
|Warnings: Please learn relative security measures and functions given in the file or on the unit.|
Don't use or store near the place which is possible to fire.
Don't use or store where air pressure or temperature or moisture exceeds range.
Keep good ventilation and avoid direct sunlight, protect against the eroding of dust and corrosive air.
Avoid slant, shaking, and shock (including transportation).
Don't use or store near to chemicals or combustible gas.
There is to indicate grounding nut, please be sure of good grounding before use.
means output of radiation.
The machine is with special device to control radiation for safe operation.
2.1 Technical specifications
tube voltage: 60KV±10%
tube current: 8mA±20%
exposal time: 0.2~4.0s
input power: <600VA
output power: P≈339VA
irradiation diameter: 6cm
fuse: RF1-20- 5A
capacity: 60KV . 8mA .
the way of operation :breake load;continuous operation
the weigh of x-ray source:7.5Kg
the variation rate of high-voltage generator 1:226
manufactory:hangzhou wandong electrical Co.Ltd.
The time of loading follows R'10
up-down range of X-ray head: 400mm
forward-back range of X-ray head: 900mm
level turning angle of head: 360°
rounding angle of head: 270°
life: 6 years
2.2 Main specifications of X-ray tube
XD2-1.4/85 fixup anode X-ray tube
|Top working voltage||Focus value||Target angle||Fuse||Perspective loading||Anode thermal capacity|
|85KV||1.5mm||23°||2.5 A||2.8±0.7V||50~60HZ||150W||9 KJ|
2.3 Main specifications of X-ray generator
|Total Al equivalent||2mmAL|