Dental Automatic X Ray Film Processor (TDH-C31)
|FOB Unit Price:||US $400 US $300|
|Purchase Qty. (Pieces)||FOB Unit Price|
|Production Capacity:||50PCS Monthly|
|Payment Terms:||T/T, Western Union, Paypal, Money Gram|
- Model NO.: TDH-C31
- Medical Devices Reg./Record No.: Gxzz20132551382
- Input Power: 4va
- Trademark: TDH
- Specification: CE ISO approved
- HS Code: 90184990
- Medical Device Regulatory Type: Type 2
- Voltage: 220V± 10V
- Atmospheric Pressure: 700hpa~1060hpa
- Transport Package: Carton Box
- Origin: China
Dental Automatic X Ray Film Processor(TDH-C31)
1). 3 steps process: Developing, fixing and bathing. The time developing films needed can be selected in the designing range. During the bathing process, delivery mechanism adopts anti-flaming ABS engineering materials. It has easy cleanliness and maintenance, stable running, small volume, operators do not touch the developers and fixers directly, convenient and safe operation and so on characteristics.
2). Main Technical parameters
A. Surroundings temperature: 10º C~30º C;
B. Relative humidity: ≤ 70%;
C. Atmospheric pressure: 700hPa~1060hPa;
D. Voltage of electrical source: 220V± 10V;
E. Frequency of electrical source: 50Hz± 1Hz.
3). Surroundings of transportation and reservation
A. Surroundings temperature range: -40º C~+55º C;
B. Relative humidity range: ≤ 93%
C. Atmospheric pressure range: 500hPa~1060hPa.
A. Processing time of film
The time which develops films needed is 2min~8min, and can be accommodated continuously. The error is within± 10%.
B. Light leaking
During the bathing process, light leaking is not permitted.
C.The stability of film delivery
The delivery mechanism of film should not lacerate or lock films.
D.The timing panel of transit film
The timing panel of transit film adopts the number 1~12 represents month, each scale represents 10 days.